The connecting rod hardness eddy current testing machine has a calibration device, input compensation parameters, automatic/manual temperature compensation and light intensity compensation, and automatic temperature measurement and temperature correction. The digital potentiometer is used to adjust the gain of the amplifier through program control. The gain parameters can be set for different battery chips, and the software working mode can be changed to adapt to various battery test applications without changing the hardware.
1. The sorting instrument is divided into hardness test (can display Brinell, Rockwell hardness), flaw detection and so on.
2. Hardness test: After starting the machine, use the hardness test probe to perform hardness test sorting on the workpiece. The Brinell or Rockwell value can be selected. The liquid crystal display will show the hardness value of the workpiece.
3. Detection of flaw detection: The workpiece to be tested shall be of the same specification. The instrument shall be calibrated with standard parts (work without damage), and the flaw detection curve on the display shall be adjusted to the upper limit and the intermediate standard point of the lower line. After calibration, Sort the workpiece for sorting.
Connecting rod condition
same size, same heat treatment process state, the same material
The form of probe
1HB Brinell, Rockwell 0.1HRC
1200 pieces per hour (specific speed depends on the size of
the workpiece under test)
Work power supply
AC 220V±10%, 50Hz
Compressed air pressure
5 — 8kgf/cm²
temperature - 10 +40, relative humidity < 7.5%.
Automation of equipment
Manual charging, automatic detection, automatic
sorting and automatic alarm
As a non-destructive testing instrument and equipment, sensor development company, BKN Technology has been an innovative manufacturer of high-quality, high-performance non-destructive testing instruments and equipment. We provide customer-centric design services to meet the different application needs of our users.
Leave us Message: